Melag Careclave 618

2025-09-12T11:57:27+01:00

Price: 10999.00

Rethink everything you know about instrument reprocessing in dental practices and clinics. Nine years of intensive research and development have brought together what belongs together: The worldwide unique 4-in-1 Solution combines the entire process of instrument reprocessing in just one device! Benefit from revolutionary advantages:

1. Space: Careclave requires up to 45 % less space, as it combines a number of devices in one.

2. Money: Reduction of running costs for maintenance and validation.

3. Time: Reprocessing and lubrication of up to 200 handpieces and further instruments in just one practice day.

The revolutionary 4-in-1 Solution for instrument reprocessing

1. Cleaning & Disinfection
Careclave guarantees the cleaning and disinfection for the reprocessing of up to 8 transmission instruments, prophylaxis handpieces or ultrasonic scaler tips in only 16 minutes.

2. Lubrication
Careclave not only ensures perfect care of your transmission instruments, but minimizes wear. And that is not all: The unique ADDcare station even performs the lubrication of the chuck system on Careclave after the program end.

3. Sterilization (Class B)
With the patented DRYtelligence and double-jacket technology, Careclave offers everything you would expect from a modern autoclave – and even more: Sterilize up to 11 kg of load on up to 8 trays or in 4 MELAstore Boxes in record time. The revolutionary chamber concept enables the reprocessing and lubrication of dental instruments in the Carebox as well as the simultaneous sterilization of other instruments on short trays in just one cycle.

4. Documentation & Approval
With ProControl you can provide proof of error-free instrument reprocessing quickly, securely and without paper work. Documentation and approval take place directly on the Careclave display without any additional software. The reprocessing protocols are stored on the MELAG USB stick or in the network for complete traceability.

Technical Data

Device dimensions (WxHxD): 48 x 50 x 65.3 cm, Height including display 56.2 cm, fits on a 60 cm table top

Chamber size: 35 cm, ø 25 cm, volume 18 l

Chamber load quantity: max. 11 kg instruments / 2 kg textiles

Electricity supply: 220-240 V, 50 Hz

Compressed air: 4-8 bar

Weight: 69 kg

Water connection: MELAdem® 47 or MELAdem® 53 required incl. USB stick, heat protection glove and side device holder for Carebox

Power consumption: 3 kW

Downloads:

Brochure (PDF)
Which Melag Autoclave is right for you? Comparison Sheet (PDF)